・Fix four locations with M2 hexagon socket head bolts.
An uneven surface affects the characteristic and may cause a breakage.
The flatness of 0.01mm or below is recommended for a mounting surface of 20mm x 20mm.
・Use a M3 screw hole at the center of a sensor to fix a specimen.
Make sure to use a screw with the length of 4mm or shorter from the pressure receiver surface (Φ6). Using a screw longer than 4.5mm may damage a sensor.
Do not overtighten a screw or it may damage a sensor since screwing applies rotation moment. Carefully tighten a M3 screw with appropriate force so that it does not become loose.
<Recommended ranges of tightening torque>
|Model||Recommended tightening torque|
|USL06-H5-50N||40N･cm or below|
|USL06-H5-100N||40N･cm or below|
|USL06-H5-200N||40N･cm or below|
|USL06-H5-500N||70N･cm or below|